Post by Lake Shore Ryan on Aug 9, 2019 14:39:23 GMT -5
Expert here , this module should be compatible. Here are the specs for the 240 Series modules, they use the DP-V0 protocol, which looks to be supported by the Prosoft MVI56 as you mentioned. Prosoft gateways like this have been used with our modules in the past too, I just don't remember which ones specifically.
Ryan Oliver Product Manager, Sensors and Instruments Lake Shore Cryotronics
I'm trying to connect Lakeshore model 240 temperature transmitter to PROFIBUS DP-V1 Master Network Interface Module for ControlLogix MVI56-PDPMV1. Unfortunately, the connection does not established. I got an error. I have now a very short (less than 1m) cable with resistor terminals on the PLC side only. When I'm trying to connect the red led of communication is flashing (parameterization stage).
In the model 240 there is only one Profibus parameter - the address. Is that enough? In the MEASURELink program, do I have to activate PROFIBUS CONNECTION button?
Can you provide online assistance on the phone? (1-240-4727650) Thanks, Artur - NIST