I am using probe station CRX-VF probe station. I read the manuall fot he probe configuration. IThe tip temperature is 45K. In such case, if I contact the tip to the sample, what is the real temperature of the sample if I set the sample stage 300K (assuming the temperature of the sample and sample stage is the same without tip)?
Is it possible to control the tip temperature or set the temperature of the tip similar to the sample stage?